# Home > Measuring-Instruments > Conductivity Meters

[ pdf ] Conductivity Measurement

``` Download 296 times Topic under  Conductivity Meters Source: globalautomation.info  File size: 190.57 KB File type: pdf Last download on: Sun Feb 18, 2018 01:49:05 PM Short Desciption: Electrolytic (Contacting) Conductivity Measurement. Introduction. All solutions containing water conduct electricity to some extent. ... Summary: All solutions containing water conduct electricity to some extent. The measure of the ability of a solution to conduct electricity is called “conductance” (the reciprocal of resistance). Addition of electrolytes such as salts, acids or bases to pure water will increase the ability of the liquid to conduct electricity. This increases the solution’s conductance (decreases the resistance). An electrolytic conductivity measuring system measures solution conductance by using an analyzer interconnected with cable to a sensor immersed in the solution. The conductivity sensor is composed of a temperature sensor and two electrodes in contact with the solution. The analyzer circuitry impresses an alternating voltage between the two electrodes and the magnitude of the resulting current is linearly related to the solution conductivity. When the temperature of a solution changes, its conductivity changes. Typically, a temperature change is compensated with a thermistor (temperature sensitive resistor) in the conductivity sensor to alter the gain of the measuring circuit. Regardless of the actual solution temperature, the display indicates what the solution conductivity would be if the solution temperature were 25°C (an internationally accepted reference). This temperature compensation may be achieved automatically or manually. All solutions do not have the same amount of conductivity change for a given temperature change. For example, the change for hydrochloric acid is 1.5% per °C, and the change for sodium chloride is 2.1% per °C. The ratio of solution conductivity change to temperature change is referred to as the temperature compensation slope and is expressed as % per °C. Bookmark or share this info on Google+ and leave comment at below : ``` ``` ```

``` ```

``` People who downloaded this also viewed in this sites: ```

What is Conductivity
All meters compensate temperature by using the world-wide accepted non-linear function ... Q. How far can my conductivity electrode be from my meter?...

Conductivity Sensors for Moored and Autonomous Operation
Conductivity (reciprocal resistivity) is an intrinsic property of ... sensor to companion packages (ALACE floats, current meters, AUVs, etc.)...

``` Leave comment :"Conductivity Measurement" ```

``` Name* Comment ```

``` Related Free Files ```

``` Conductivity Sensors for Moored and... - Download: 171 timesWhat is Conductivity... - Download: 251 times ```

1
at 0 : 6 : 3 ago
Topic: HVAC System

2
at 0 : 7 : 30 ago
Topic: HVAC System

3
at 1 : 41 : 6 ago

4
at 1 : 49 : 34 ago
Topic: Process Control

5
at 2 : 2 : 10 ago
Topic: Microcontrollers

6
at 2 : 47 : 34 ago
Topic: CNC

7
at 3 : 10 : 5 ago
Topic: API

8
at 4 : 10 : 51 ago
Topic: PLC (Programmable Logic Controller)

9
at 8 : 51 : 14 ago

10
at 9 : 51 : 44 ago
Topic: HVAC System

11
at 9 : 52 : 20 ago
Topic: HVAC System

12
at 10 : 29 : 42 ago
Topic: Rotary actuator

13
at 10 : 31 : 0 ago
Topic: Electronics Systems

14
at 11 : 14 : 45 ago
Topic: Machine Systems

15
at 13 : 27 : 48 ago
Topic: Automation