# Home > Measuring Instruments > Flow Measurement

```
```

**Download**

**64**

times
**Topic under **Flow Measurement

**Source:** www.mscsoftware.com

**File size:** **834 KB**

**File type:** **pdf**

**Last download on:**
Fri Apr 20, 2018 04:23:34 AM

__Short Desciption__:

A Coriolis mass ilowmeter measures mass flow directly by vibrating a fluid—conveying pipe
at resonance. Design of these meters was facilitated by solving the equations of motion via
the finite element method. A complex eigenvalue analysis (SOLIU7) was necessary due to the
presence of the Coriolis force terms in the damping matrix. Performance parameters of the
meter can be predicted from the complex eigenvectors and were found to match experimentally
determined values to within five percent
Since vibration occurs during Coriolis meter operation, low stress designs must be used to
preclude fatigue failure. This analysis was also performed using MSC/NASTRAN

__Summary__:

MSC/NASTRAN is used extensively in the design and analysis of Coriolis mass iiowmeters. The
early portion of the design cycle focuses on the prediction of meter dynamics. MSC/NASTRAN
was used at this stage to solve (SOLlO7) the equations of motion of a custom beam finite element.
This finite element was imported into MSC/NASTRAN via DMIG entries. Additionally the linear
algebra capabilities of DMAP were used to allow the engineer to analyze an entire spectrum of A
mass fiow rates and fiuid process densities with a single finite element model.
Standard MSC/NASTRAN elements and solutions were also used at various points in the design
cycle to estimate stress. Estimation of stress is necessary to preclude fatigue failure. The ability
to predict both meter dynamics and stress in software significantly shortens the development
time for new Coriolis mass fiowmeters.

Bookmark or share this info on Google+ and leave comment at below :

```
```

```
```

```
People who downloaded this also viewed in this sites:
```

__STANDARD HANDBOOK OF PETROLEUM AND NATURAL GAS ENGINEERING Chp 1 to 3__

Petroleum engineering now has its own true classic handbook that reflects the profession’s status as a mature major engineering discipline.
...__Flow Measurement__

respiratory research the most widely used flow measurement device is the Fleisch type differential pressure with a heated screen orifice....__ORIFICE METERING FUNDAMENTALS __

THEORY OF ORIFICE MEASUREMENT
The law of conservation of energy applied to a fluid under
the condition of steady, uniform flow requires the...

```
Leave comment :
```

"DESIGN AND ANALYSIS OF CORIOLIS MASS FLOWMETERS USING MSC NASTRAN"

```
Related Free Files
```

```
STANDARD HANDBOOK OF PETROLEUM AND NATURAL... - Download:
```

**621** times

ORIFICE METER PRIMARY ELEMENTS STANDARDS... - Download: **70** times

Measurement of flowing fluids... - Download: **79** times

Flow Orifice Sizing... - Download: **83** times

Guide for Flow Measurement... - Download: **208** times

Flow Handbook... - Download: **551** times

Measurement of gas and liquids by orifice... - Download: **403** times

FUNDAMENTALS OF ORIFICE RECORDERS... - Download: **32** times

Proving Coriolis Flowmeters... - Download: **64** times

DESIGN AND ANALYSIS OF CORIOLIS MASS... - Download: **26** times