Vortex Flow Meters
File size: 864.66 KB
File type: pdf
Last download on: Sat Mar 25, 2017 02:32:05 PM
The vortex shedding flowmeter first emerged 25 to 30 years ago and has steadily grown in acceptance since then to be a major flow measurement technique. Its appeal is due, in part, to the fact that it has no moving parts yet produces a frequency output that varies linearly with flowrate over a wide range of Reynold’s numbers. The vortex meter has a very simple construction, provides accuracy (1% or better) comparable to higher price and/or more maintenance intensive techniques and works equally well on liquids and gases. In addition it is powered primarily by the fluid and lends itself more readily than other linear flow devices to two -wire operation. Comparing the vortex shedding flow meter to an orifice plate the former has higher accuracy and range ability, does not require complex pressure impulse lines, is less sensitive to wear and, for volumetric flow measurement, does not require the need to compensate for fluid density.
What’s Inside: 1. Introduction 2. Principle of Operation 3. Calculation of Mass Flow and Standard Volume 4. Flowmeter Construction 4.1 Flowtube 4.2 Transmitter 5. Application Considerations 5.1 Meter Selection 5.2 Meter Installation 5.3 Meter Configuration 6. Recent Developments 6.1 Multi-sensor 6.2 Single-sensor 7. Additional Reading
Bookmark or share this info on Google+ and leave comment at below :
People who downloaded this also viewed in this sites:
Frequency of shedding vortices in the wake of a bluff body is used to measure the flow rate in vortex flowmeters. A smooth bluff body shape may...
Leave comment :
"Vortex Shedding Tutorial"
Related Free Files
Effect of bluff body shape on vortex flow... - Download: 87 times